|Portable Current Signal Generator
Portable Current Signal Generator
* Products that meet your criteria are shown in
- This product is manufactured in Japanese standards. Please check the specification before you order.
- Fixed Output
- Can be set fixed output of 4, 12, 20mA in common / (fixed value) VR output change-over switch
- Opreration Mode Switch
- The switchable 4-wire and 2-wire at the 2-wire / 4-wire selector switch
- Maximum Input Voltage
- 2-wire mode / nominal DC24V (DC28V or less)
- Monitor Output
- Common / monitor selector switch can be monitored at the time of the M side
- Load Resistance
- 4-wire mode / (acceptable) 750ohm
- LED Color
- Red (POWER switch ON lighting)
- Output Range
- The output can be set to 24mA - DC0 in common / (VR) analog volume adjustment
- Material (Housing)
- ABS resin
- Continuous Opreration Time (hours)
- About 7 (load resistance 750ohm, 12mA output, new AAA alkaline batteries 2)
- test plug of phi2
- The direct measurement monitor / output current
- Voltage (V)
- 4-wire mode / (maximum output terminals) DC21.5 +/-0.5
- Hole Diameter (phi mm)
- Monitor terminal / phi2
- Use form / stationary stand, strap mounting holes
- Test leads (1.2m), AA batteries (two)
Customers Who Viewed This Item Also Viewed
GE Sensing & Inspection Technologies
SANWA ELECTRIC INSTRUMENT
| 5 star
| 4 star
| 3 star
| 2 star
| 1 star
5 Customer Reviews
The reviews are automatic translation.
Operation test of the sequencer
On 2016-07-07 17:02:29
It is enough if you do not ask for high precision.
On 2015-12-15 10:32:00
It is more compact and light than I thought. Easy to use, you can work well without instruction manual.
It is enough to use one for checking if the current value is measured even if it is broken.
On 2015-11-20 16:40:36
Those who can understand the concept are good, but it is good to have materials so that you can understand 2 wire and 4 wire.
On 2015-09-24 23:57:05
I borrowed it and used it, it was convenient so I bought it. It is used for IO checking and debugging of the instrumentation program, but it is quite nice to use it with two smaller batteries. The price is expensive, but it is really recommended if you do not need as much features as a more expensive calibrator. I think that it was okay for me to just turn on the voltage output. It only adds 250 ohms or 500 ohms of resistance ...
Test drive of electro-pneumatic converter
On 2014-02-13 15:56:35
It is compact and easy to carry, but I think that it would be better if the installation could be done in the box Output format was also convenient as various patterns could be selected